Electromagnetic sensors based on magnonic crystals for applications in the fields of biomedical and NDT
|Author(s)||Talbot Ph1, Fessant A.2, Gieraltowski J.3|
|Affiliation(s)||1 : Univ Bretagne Occidentale, Lab STICC, UMR CNRS 6285, CS 93837,6 Ave Gorgeu, F-29238 Brest 3, France.
2 : Univ Bretagne Occidentale, LMB, F-29238 Brest 3, France.
3 : Univ Bretagne Occidentale, LDO IUEM UMR CNRS 6538, F-29238 Brest 3, France.
|Meeting||Conference on EUROSENSORS 2015, Freiberg, GERMANY, SEP 06-09, 2015|
|Source||Procedia Engineering (1877-7058) (Elsevier Science Bv), 2015 , Vol. 120 , P. 1241-1244|
|WOS© Times Cited||8|
|Keyword(s)||Magnonic crystals, Magnetic field sensor, YIG film, Magnetostatic surface wave (MSSW), Band gap|
An exploratory investigation of high sensitivity sensors based on magnonic crystals for the measurement of weak magnetic fields at room temperature is presented. The samples are YIG crystals on GGG substrate (Gadolinium -Gallium -Garnet) on which a periodic structure of shallow grooves are etched or gold stripes are deposited. The excitation of surface magnetostatic spin waves (MSSW) which should appear in these structures is obtained by means of microstrip transmission line or microstrip planar antenna. The measurement of the magnonic structure by complex transmission coefficient (S12) shows a significant shift in frequency of the maximum absorption peak related to magnonic band gap which depends on the magnitude of the DC magnetic bias field applied. It shows the possibility of magnonic high Q -value band gap implementation with respect to spin wave propagation band (GHz band) in magnonic crystals applied to spin waves detection of magnetic fields. The influence of the characteristics of the microstrip line or the microstrip antenna on sensor performance (sensitivity and resolution) is studied. (C) 2015 article under the CC BY-NC-ND license (http://creativecommons.org/licenses/by-ric-nd/4.0/) Peer-review under responsibility of the organizing committee of EUROSENSORS